Payment & Shipping Terms:
|Measurement:||Tension/Compression||Type:||S Type Load Cell|
S Beam Load Cell 1.1klb 2.2klb 4.4klb 6.6klb 8.8klb 11klb 16.5klb S Type Force Sensor
High Accuracy S Type Load Cell FSRT is a compact stainless steel load cell suitable for both tension and compression load measurement. S Beam Force Sensor FSRT offers 0-1.1klbf / 0-2.2klbf / 0-4.4klb / 0-6.6klb / 0-8.8klb / 0-11klbf / 0-13klb / 0-16.5klb eight rated measuring ranges to choose from. Unified 2.0mV/V output,high accuracy with 0.02% combined error of the rated capacity. Ideal for harsh environments, it’s fully welded stainless steel and sealed to IP68. Suitable for asphalt, concrete, and aggregate batching plants. Other popular uses include power line tensioning, underground cable pulling and hanging tanks, hopper scales.
500kg / 5kN / 1.1klbf
1000kg / 10kN / 2.2klbf
2000kg / 20kN / 4.4klbf
3000kg / 30kN / 6.6klbf
4000kg / 40kN / 8.8klbf
5000kg / 50kN / 11klbf
6000kg / 60kN / 13klbf
7500kg / 75kN / 16.5klbf
|Rated output||2.0 mV/V||Compensated temperature||-10...+40oC|
|Zero balance||±0.03mV/V||Temperature shift zero||±0.02% R.O./10oC|
|Nonlinearity||±0.017% R.O.||Temperature shift span||±0.02% R.O./10oC|
|Hysteresis||±0.017% R.O.||Input resistance||350±5 ohms|
|Nonrepeatability||±0.01% R.O.||Output resistance||350±3 ohms|
|Creep(30min)||±0.017% R.O.||Insulation resistance||>5000Mohms|
|Safe overload||150% F.S.||Ingress protection||IP68|
|Ultimate overload||300% F.S.||Material of element||Stainless steel|
|Cable||Ø5*6000mm 4-core shielded|
Red--Power input+ Black--Power input-
Green--Signal output+ White--Signal output-
Load cell glossary:
Load Cell:A device which produces an output signal proportional to the applied weight or force.
Wheatstone Bridge:A Wheatstone bridge is a network of four resistive legs. One or more of these legs can be active sensing elements. The Wheatstone bridge is the electrical equivalent of two parallel voltage divider circuits. R1 and R2 compose one voltage divider circuit, and R4 and R3 compose the second voltage divider circuit. The output of a Wheatstone bridge is measured between the middle nodes of the two voltage dividers.
Excitation: The voltage applied to the input terminals of the load cell.
Rated Capacity: The maximum axial load the load cell is designed to measure within its specifications.
Non-linearity: The maximum deviation of the calibration curve from a straight line drawn between the no-load and rated outputs; expressed as a percentage of the rated output and measured on increasing load only.
Non-repeatability: The maximum difference between load cell output readings for repeated loadings under identical loading and environmental conditions.
Hysteresis: The maximum difference between load cell output readings for the same applied load; one reading obtained by increasing the load from zero and the other by decreasing the load from rated output; usually measured at half rated output and expressed in percent of rated output.
Creep: The change in load cell output occurring with specified period of time, while under load, and with all environmental conditions and other variables remaining constant; usually measured with rated load applied.
Safe Overload: The maximum load in percent of rated capacity which can be applied without producing a permanent shift in performance characteristics beyond those specified.
Ultimate Overload: The maximum load in percent of rated capacity which can be applied without producing a structural failure.
Operating Temperature: The extremes of temperature within which a load cell will operate without permanent adverse change to any of its performance characteristics.
Temperature Shift Span: The change in output due to a change in load cell temperature; usually expressed as a percentage of load per degree Celsius change in temperature.
Temperature Shift Zero: The change in zero balance due to a change in ambient temperature; usually expressed in percent of rated output per degree Celsius change in temperature.
Input Resistance: The resistance of the load cell bridge circuit measured at the positive and negative excitation terminals with no load applied and with the output terminals open-circuited.
Output Resistance: The resistance of the transducer bridge circuit measured at the positive and negative signal terminals with no load applied and with the input terminals open-circuited.
S Beam Force Transducer 1.1klb 2.2klb 4.4klb 6.6klb 8.8klb 11klb 16.5klb S Type Load Cell