Payment & Shipping Terms:
|Usage:||Compression Tension Force Measurement||Style:||Universal S Beam Load Cell|
|Output:||Analog Sensor||Load Direction:||Tension And Compression|
|Capacity:||50N 100N 200N 300N 500N 1kN 2kN 3kN||Threaded Hole:||M8/M12|
Universal S Type Tension Load Cell 3kN 2kN 1kN 500N 300N 200N 100N 50N
Universal s-beam Tension Load Cell FS6A/FS6B is strain gauge based intended for tension and compression force measurement. S Type Load Cell FS6A/FS6B offers 0-50N / 0-100N / 0-200N / 0-300N / 0-500N / 0-1000N / 0-2000N / 0-3000N eight rated measuring ranges to choose from. M8/M12 metric threaded holes located on both ends can be used to mount attachments such as eye bolts and hooks for adapt to different cases. S Type Force Transducer FS6A/FS6B offers an accurate and affordable measuring solution for applications such as material testing machines, hanging scales, hopper weighing, automation equipment and so on.
5kg / 50N / 11 lb
10kg / 100N / 22lbf
20kg / 200N / 44lb
30kg / 300N / 66 lbf
50kg / 500N / 110lbf
100kg / 1kN / 220 lb
200kg / 2kN / 440lbf
300kg / 3kN / 660 lb
|Safe overload||150% F.S.|
|Ultimate overload||200% F.S.|
|Temperature shift zero||±0.05% R.O./10oC|
|Temperature shift span||±0.05% R.O./10oC|
|Input resistance||385±50 ohms|
|Output resistance||380±30 ohms|
|Insulation resistance||>5000M ohms|
|Material of element||Alloy steel|
|Cable||Ø5.5*2000mm 4-core shielded|
Load cell glossary:
Wheatstone Bridge:A Wheatstone bridge is a network of four resistive legs. One or more of these legs can be active sensing elements. The Wheatstone bridge is the electrical equivalent of two parallel voltage divider circuits. R1 and R2 compose one voltage divider circuit, and R4 and R3 compose the second voltage divider circuit. The output of a Wheatstone bridge is measured between the middle nodes of the two voltage dividers.
Excitation: The voltage applied to the input terminals of the load cell.
Rated Capacity: The maximum axial load the load cell is designed to measure within its specifications.
Non-linearity: The maximum deviation of the calibration curve from a straight line drawn between the no-load and rated outputs; expressed as a percentage of the rated output and measured on increasing load only.
Non-repeatability: The maximum difference between load cell output readings for repeated loadings under identical loading and environmental conditions.
Hysteresis: The maximum difference between load cell output readings for the same applied load; one reading obtained by increasing the load from zero and the other by decreasing the load from rated output; usually measured at half rated output and expressed in percent of rated output.
Creep: The change in load cell output occurring with specified period of time, while under load, and with all environmental conditions and other variables remaining constant; usually measured with rated load applied.
Safe Overload: The maximum load in percent of rated capacity which can be applied without producing a permanent shift in performance characteristics beyond those specified.
Ultimate Overload: The maximum load in percent of rated capacity which can be applied without producing a structural failure.
Operating Temperature: The extremes of temperature within which a load cell will operate without permanent adverse change to any of its performance characteristics.
Temperature Shift Span: The change in output due to a change in load cell temperature; usually expressed as a percentage of load per degree Celsius change in temperature.
Temperature Shift Zero: The change in zero balance due to a change in ambient temperature; usually expressed in percent of rated output per degree Celsius change in temperature.
Input Resistance: The resistance of the load cell bridge circuit measured at the positive and negative excitation terminals with no load applied and with the output terminals open-circuited.
Output Resistance: The resistance of the transducer bridge circuit measured at the positive and negative signal terminals with no load applied and with the input terminals open-circuited.
Universal S Beam Tension Load Cell Force Transducer 3kN 2kN 1kN 500N 300N 200N 100N 50N
Contact Person: Mr. Tarik Lu
Tel: +86 755 84536383