Payment & Shipping Terms:
|Usage:||Load Measurement||Type:||Planar Beam Load Cell|
|Output:||Analog Sensor||Load Direction:||Compression|
Low Profile Planar Beam Load Cell 300kg 150kg 75kg 30kg Weight Transducer
Low Profile Compression Load Cell FHA is produced based on strain gauge technology with CE certified. Low profile compression load cell FHA offers 0-30kg / 0-75kg / 0-150kg / 0-300kg four measuring ranges to choose form, high accuracy with 0.05% non-linearity of the full capacity. Thin load cell FHA utilizes 1000ohms gauges which has better performance compared to commonly used 350ohms gauges in terms of power consumption and long term stability. Compact design enables the construction of scales with an extremely low overall height, affordable load cell FHA is also suitable for battery powered weighing devices.
30kg / 300N / 66 lb
75kg / 750N / 165lb
150kg / 1500N / 330 lb
300kg / 3000N / 660lb
|Rated output||1.5/2.0 mV/V|
|Safe overload||150% F.S.|
|Ultimate overload||200% F.S.|
|Temperature shift zero||±0.2% R.O./oC|
|Temperature shift span||±0.05% R.O./10oC|
|Input resistance||1120±50 ohms|
|Output resistance||1000±10 ohms|
|Insulation resistance||>2000M ohms|
|Material of element||Aluminum alloy|
|Cable||30kg/75kg — 32AWG 400mm 4-wire
150kg/300kg — 32AWG 150mm 4-wire
A load cell is a transducer that converts mechanical load into mV signal proportional to the applied load/force.The acting force causes deformation of the measuring element,the bonded strain gage deforms accordingly which will cause the resistance change of the wheatstone bridge,under constant power supply, the output signal between Sig+ and Sig- will change proportionally,thus the load can be measured by measuring the output signal from the transducer which is normally within -30mV...+30mV.
Brief instruction of Rated output(or Sensitivity):
Definition: The signal produced by load cell(or force transducer) when rated load is applied, expressed in terms of mV/V. The rated output indicates the output signal range of the sensor.
Sensor output = (Power supply * Rated output) * Load / Capacity
Rated output is normally within 1mV/V~2mV/V.
Power supply is normally within 3-15V.
Load / Capacity is within 0-1.
So the load cell output signal is normally within 0-30mV(-30mV...+30mV for bidirectional load cell).
Example one: FHA-30kg with 1.5mV/V rated output powered by 3.3V DC ,the output range is 0~4.95mV.
Example two: FHA-300kg with 2mV/V rated output powered by 5V DC ,the output range is 0~10mV.
Low Profile Load Cell 3kN 1.5kN 750N 300N Compression Force Sensor
Contact Person: Mr. Tarik Lu
Tel: +86 755 84536383