Payment & Shipping Terms:
|Usage:||Compression Force Measurement||Type:||Low Profile Load Cell|
Low profile compression load cell 100kg Ultra thin force transducer 1000N
Low profile compression load cell FMXD utilizes strain gauge technology which are bonded on to the alloy steel sensing body.Ultra thin load cell FMXD is available in 0~10kg / 0~30kg / 0~50kg / 0~100kg four measuring ranges with maximum 0.1% F.S. non-linearity, it offers an accurate & low cost measurement solution for high volume applications.The 5.1mm thru hole located in the center of load cell can be used to mounting attachments to adapt to different applications. With its compact design, transducer FMXD low profile force sensor is suitable for applications where vertical space is limited,it is ideal for use in robotics,process control and automotive applications,etc..Custom made service is available for other measuring ranges with similar structure.
100N / 10kg / 20 lb
1000N / 100kg / 220 lb
|Safe overload||120% F.S.|
|Ultimate overload||150% F.S.|
|Temperature shift zero||±0.03% R.O./oC|
|Temperature shift span||±0.03% R.O./oC|
|Input resistance||1000±100 ohms|
|Output resistance||1000±100 ohms|
|Insulation resistance||>2000M ohms|
|Material of element||Alloy steel|
Load cell glossary:
Load Cell:A device which produces an output signal proportional to the applied weight or force.
Wheatstone Bridge:A Wheatstone bridge is a network of four resistive legs. One or more of these legs can be active sensing elements. The Wheatstone bridge is the electrical equivalent of two parallel voltage divider circuits. R1 and R2 compose one voltage divider circuit, and R4 and R3 compose the second voltage divider circuit. The output of a Wheatstone bridge is measured between the middle nodes of the two voltage dividers.
Excitation: The voltage applied to the input terminals of the load cell.
Rated Capacity: The maximum axial load the load cell is designed to measure within its specifications.
Non-linearity: The maximum deviation of the calibration curve from a straight line drawn between the no-load and rated outputs; expressed as a percentage of the rated output and measured on increasing load only.
Non-repeatability: The maximum difference between load cell output readings for repeated loadings under identical loading and environmental conditions.
Hysteresis: The maximum difference between load cell output readings for the same applied load; one reading obtained by increasing the load from zero and the other by decreasing the load from rated output; usually measured at half rated output and expressed in percent of rated output.
Creep: The change in load cell output occurring with specified period of time, while under load, and with all environmental conditions and other variables remaining constant; usually measured with rated load applied.
Safe Overload: The maximum load in percent of rated capacity which can be applied without producing a permanent shift in performance characteristics beyond those specified.
Ultimate Overload: The maximum load in percent of rated capacity which can be applied without producing a structural failure.
Operating Temperature: The extremes of temperature within which a load cell will operate without permanent adverse change to any of its performance characteristics.
Temperature Shift Span: The change in output due to a change in load cell temperature; usually expressed as a percentage of load per degree Celsius change in temperature.
Temperature Shift Zero: The change in zero balance due to a change in ambient temperature; usually expressed in percent of rated output per degree Celsius change in temperature.
Low profile compression load cell 100kg Ultra thin force sensor 1000N
Contact Person: Mr. Tarik Lu
Tel: +86 755 84536383